IN the aftermath of an aboveground nuclear explosion, measuring the isotopic compositions of fallout can provide important forensic information, including data pertaining to the production and irradiation histories of key materials. Livermore scientists working with collaborators at Argonne National Laboratory and the University of California at Berkeley have recently demonstrated rapid and high-accuracy isotopic measurements of plutonium and uranium using a laser-based technique known as resonance ionization mass spectrometry (RIMS).
Resonance ionization uses pulsed laser light to first excite an atom or molecule and then remove an electron to create an ion. Although scientists have understood the basic physical principles involved in the process for many years, it was not until the mid-1990s that a method was developed to provide unprecedented analysis of stellar nucleosynthesis processes captured in grains of star dust. Lawrence Livermore researcher Kim Knight, whose background is in cosmochemistry, and colleagues theorized that the RIMS technique could also be applied to study samples derived from nuclear events on Earth.
Postdetonation nuclear forensics is primarily concerned with detecting the presence of uranium (U) and plutonium (Pu) and measuring isotopic ratios, 235U/238U for example. Conventional mass spectroscopy cannot distinguish between different isotopes of identical mass—in particular, 238U and 238Pu—in the same sample. As a result, the two actinides must be extracted from a forensic sample and separated from each other in a process that can take longer than decision makers would want to wait. The challenge is that postdetonation nuclear forensics demands rapid answers. RIMS uses lasers to separate elements, thus circumventing the lengthy steps of chemical separation and sample preparation. As a result, several actinides can be measured from virtually unprocessed samples in only a few hours.
Calming the Jitters
Building lasers for demanding forensics applications has been a significant challenge. Even tiny fluctuations in laser wavelength, power, and bandwidth can introduce errors that affect the reproducibility of results. According to Knight and Argonne collaborator Michael Savina, stabilizing the lasers and reducing jitter has greatly improved the quality of the measurements. By precisely controlling wavelength, pulse timing, power, and pointing stability in the lasers and making adjustments to bandwidth (the spectral range of laser wavelengths present in the laser beam), the team achieved more uniform ionization in experiments and thus better detection and measurement of multiple isotopes.
Speeding Up the Process
Variability in the measured isotope ratio means that each isotope has responded differently to the ionization process and can lead to incorrect conclusions. By broadening the laser bandwidth from 1 to 5 picometers (10–12 meters) in the first of three resonance lasers, while holding the second and third resonance lasers fixed, the team decreased measurement uncertainty from 10 percent to less than 0.5 percent. The tiny adjustment works because the 5-picometer-wide range of laser wavelengths overlaps the peak energy region needed to excite both 235U and 238U with equal intensity. This study shows that laser bandwidth has an enormous effect on the precision of isotope ratio measurements and demonstrates how to optimize that particular parameter.
In a subsequent study, the team completed the first in situ analysis of a complex natural uranium silicate (uranium ore) with RIMS using seven experimental configurations. Both two- and three-color laser schemes were assessed along with different methods for dislodging uranium atoms from a sample. To improve precision, the team used an automatic feedback system to track and correct drift in the laser wavelengths and broadened the laser bandwidth to as wide as 10 picometers. Results were achieved in a few hours with precisions of 1 percent for uranium isotope compositions measured directly from the uranium silicate after minimal sample preparation.
To measure both uranium and plutonium isotopes in a single material for the first time, the RIMS team prepared samples from a solution of 25 percent each of 238Pu, 239Pu, 242Pu, and 244Pu that also contained 235U and 238U. Sample material was electrodeposited onto a titanium stub to promote reduction. The CHARISMA spectrometer was used with lasers tuned for a three-color excitation and ionization. The challenge was that 238U, which is the most common uranium isotope in nature, creates isobaric (same-mass) interference with plutonium, making the two isotopes difficult to differentiate. By making improvements to the excitation lasers and changing the laser wavelengths between measurements, the team demonstrated excellent discrimination against isobaric interferences. When the lasers were tuned to excite plutonium, the uranium signal was essentially absent, and inversely, the plutonium signal was essentially absent when lasers were tuned to excite uranium. All of the plutonium and uranium isotope ratios were accurately measured.
Inside Desert Glass
The team’s research findings demonstrate how RIMS can be adapted to measure different elements in a sample and discriminate against interfering masses. As a result, RIMS holds considerable promise for applications in nuclear forensics. Knight and her colleagues continue to explore ways to improve stability and reproducibility in their measurements. Future studies may include developing techniques to increase the amount of atomic—as opposed to molecular—material in the cloud, suppress molecule ionization, and increase sensitivity to desired elements, improving detection in low-concentration materials.
Key Words: Chicago-Argonne Resonance Ionization Spectrometer for MicroAnalysis (CHARISMA), fallout debris, nuclear forensics, plutonium isotope, resonance ionization mass spectrometry (RIMS), uranium isotope.
For further information contact Kim Knight (925) 422-9396 (email@example.com).
Lawrence Livermore National Laboratory
Privacy & Legal Notice | UCRL-TR-52000-12-4/5 | April 6, 2012